Completed with IPN-01 or IPN-01М
|Piezoelectric vibration transducer||AR62V||
Completed with MCC-01 or MCC-01М
|Piezoelectric vibration transducer||AR63V|
|Measurement rangers :|
|RMS of voltage, mV||Up to 1000|
|RMS of charge, pC||Up to 1000|
|RMS of vibration acceleration, m/s2||Up to 100|
|RMS of vibration velocity, mm/s||Up to 32|
|Maximum frequency range:|
|vibration acceleration, voltage, charge, Hz||10–10 000|
|vibration velocity, Hz||10–2500|
|Cut-off frequency of digital filter, Hz||1000, 2500, 5000|
|Relative error limits|
|measuring transducer with vibration transducer, %||±6|
|measuring transducer, %||±3|
|Interface||RS-485, Modbus RTU protocol|
|Rate of exchange, kB/s||1,2–115,2|
|Digit capacity of analogue-to digital converter, bit||12|
|Sampling frequency, kHz||25|
|Power supply voltage, V||18–36|
|Consumed power, W, at least||2|
|Maximum useful current, mA, at least||75|
|Galvanic isolation of power supply . . . . . . . . . . . . . From output circuits and housing|
|Mean time to failure, h, at least||30 000|
|Average service life, years||12|
|Operating frequency range, °С:|
|measuring transducer primary transducers:||-40 ... +70|
|AR36||-40 ... +125|
|AR62||-60 ... +250|
|AR63||-60 ... +400|
|Explosion-proofness according to GOST 14254-96:|
|measuring transducers MVC-01 and MCC-01||IP54|
|measuring transducers MVC-01М and MCC-01М||IP30|
Operation details: State Register No. 34095-07. Type Approval Certificate RU.C.28.011.A No. 26984. State Register of the Republic of Belarus No. RB 03 06 3544 07. Type Approval Certificate No. 4925.
It operates at GAZPROM facilities since 2004. Kirovenergomash, Vega-GAZ, Borisov Plant of Aggregates (the Republic of Belarus), Volzhsk Electric Company, Rostov NPP.
Scope of delivery: absolute vibration measurement equipment IVA-I ITsFR.402248.003, passport ITsFR.402248.003PS, operational manual ITsFR.402248.003RE, CD with software and operational documentation.
Single values of measured parameters are not specified if the modes corresponding to them are not used.
A base exchange protocol via interface RS-485 – MODICON MODBUS RTU in compliance with Modicon Modbus Reference Guide (PI-MBUS-300 Rev.C). If it is necessary to use an exchange protocol that differs from the base one, please, specify its distinctive features while making an order.
Ordering information: ((it is possible to use an abbreviated notation): Equipment IVA-I ITsFR.402248.003 -IPO-VPO/5-50/32/X-UI-F1:
Absolute vibration measurement equipment IVA-I ITsFR.402248.003 as part of a measuringtransducer IPN-01 completed with a vibration transducer AR36-100-01 with the cable’s length of 5 meters, measurement range of vibration acceleration RMS: 0,5–50 m/s2, vibration velocity RMS: 0,5–32 mm/s., with set up DC current outputs 4–20 mA and 0–5 V voltage, a programmable switchable filter with 1000 Hz cut-off frequency, a frequency measurement mode is not set up (Х symbol), exchange protocol MODBUS RTU.
Equipment IVA-I ITsFR.402248.003 – IP1-VPKh/Х-500/250/1500-I-F2:
Absolute vibration measurement equipment IVA-I ITsFR.402248.003 as part of measuring transducer IPZ-01, measurement range of charge RMS: 2,5–500 pC, range of a charge RMS on base frequency – 159.2 Hz for integration and measurement of integrated charge RMS – 1,25–250 pC, frequency:10–1500 Hz,, with set up DC current outputs 4–20 mA, a programmable switchable filter with cut-off frequency 2500 Hz, exchange protocol MODBUS RTU.
|1||GNA – internal shield of vibration transducer’s cable||1||+Usup|
|2||–Utr||2||RS-A – digital output|
|3||IN– – input||3||RS-В – digital output|
|4||IN+ – input||4||GND – general|
|5||+Utr||5||U – voltage output|
|6||+5V||6||I – DC current output|
|1||+Ubuffer – voltage output||1||+Usup|
|2||-Ubuffer – voltage output||2||–Usup|
|3||+Utr (input IPN-01М)||3||RS-A – digital output|
|4||IN+ – input||4||RS-B – digital output|
|5||input||5||GND – general|
|6||–Utr (input IPN-01М))||6||I – dc current output|
|7||GNA – internal shield of vibration transducer's cable||7||U – voltage output|
|8||GN – grounding of vibration transducer's housing||8||SYS – discrete output|
Notes. 1. Circuits +5V and –5V – for power supply to primary transducers with independent signal and power supply circuits.
2. Circuits +Utr and –Utr – for power supply to primary transducers via a signal circuit (via 750 Ohm resistors from +5V and –5V circuits respectively).
3. +Utr and –Utr contacts in a measuring transducer MCC-01М are not available.
4. Circuits +Utr with Input IN+ and –Utr with Input IN– of a measuring transducer MVC-01 and vibration transducer AR36-100 are connected using jumpers in the vibration transducer's connector.
3. Measuring transducer IPZ-01
|Input connector||Output connector|
|1||+Usup||1||GNA – internal shield of vibration transducer’s cable|
|2||RS-A – digital output||2||GN – vibration transducer grounding|
|3||RS-B – digital output||3||IN– – input|
|4||GND – general||4||IN+ – input|
|5||U – voltage output|
|6||I – DC current output|